CONTENTS. Service Manual ECG C Contents The CE mark is applied only to the ECGL/M/N Electrocardiograph. This equipment complies. Service Manual ECG Composition. · To order a replacement assembly above, use the Code No. · To order a replacement component inside an. View Homework Help – Nihon_Koden_Cardiofax_ECG_-_Repair_manual from BA at Vietnam National University, Ho Chi Minh City. ECGL.
|Published (Last):||26 August 2014|
|PDF File Size:||1.85 Mb|
|ePub File Size:||3.7 Mb|
|Price:||Free* [*Free Regsitration Required]|
To achieve the goal of “co-win”, we provide three way of labels for customers, one is our own neutral label with no logo on each battery, two is with our “COWON” brand logo on each battery, three is with customer’s own logo so that people can tell the brand at a glance, enhance the publicity.
Contact Supplier Start Order. 9620 the paper sensor.
No interrupt signal of ms. Removing the Magazine and Recording Paper The cardiograph may malfunction due to high-frequency noise from the ESU.
Refer to the following explanations for each test. To re-enter the system test number, do either of the following: Call up the System Test Level 2. Local language is not installed. Do not leave the battery pack unused for more than one year. Power cord, patient cable are not frayed and are correctly connected to the instrument. Never damage or directly solder the sheath tube.
All settings are correct. Do not immerse the battery pack in water or seawater. Printout of Test 920l 1 Not printed in Test level 2.
The battery is not charged. Meaning Corrective Action 00 01 02 03 04 05 06 07 09 Input unit error: Replace the power fuse with a new one.
Because the outside surface of the instrument is made of resin, the outside surface of the instrument is easily damaged. Use only parts recommended by Nihon Kohden to assure maximum performance from your instrument. Dirty motor 6920l sensor. The power transformer is faulty.
Remove the thermal head cable from the thermal head. Replace the thermal head. Test level 1 for operator and Test level 2 for qualified service personnel. Diagonal lines are displayed.
ECG Machines For Sale
The lithium battery to back up the date and time and all system settings is completely discharged. Before defibrillation, all persons must keep clear of the bed and must not touch the patient or any equipment connected to the patient. Check that the electrodes are properly attached to the patient. ECG interpretation error Time over. Following is the LCD display after the system has been initialized. Before defibrillation, remove all electrodes and gel from the chest of the patient.
The instrument must not be altered or modified in any way.
Before You Begin Removing, replacing and installing major components should be done by qualified service personnel. Never store or carry the battery pack with metal such as necklace or hair pins. Remove the three M3 binding head screws which fasten the thermal head unit to the lower casing and remove the thermal head unit. Recording resolution setting1 Recording resolution setting2 Recording resolution setting3 Recording resolution setting4 Recording resolution setting5 Recording resolution ecgg Recording resolution setting7 Recording resolution setting8 Recording resolution setting          Label: Replace the key board.